Belmed F250 Portable Dental Flowmeter PC-7 With Tall Mobile Stand + Scavenger
If you are looking for a fully mobile system with tanks attached to the system, see the Belmed F300 Page. This particular unit is called the Belmed F250 and is a portable dental flowmeter system designed to be used with wall outlets or stationary tanks, usually connected to 5 feet of N2O2 and O2 hoses. So if you have wall outlets already installed, this is a good system for you, if not then the Belmed F300 is your best option. The F250 comes with a scavenging circuit, breathing circuit, Nitrous and Oxygen flowmeter head, tall stand with wheels, and the nasal hood.
WARNING: This product can expose you to chemicals including arsenic, which is known to the State of California to cause cancer. For more information, go to www.P65Warnings.ca.gov.
For questions or quotes
- Additional information
Belmed F250 Flowmeter Characteristics
Flowmeter Head Dimensions – 8.75″H X 3.75″W X 10.25″D
Weight of Flowmeter Head – 5 LBS
Bag and Outlet Diameter – 22MM (APPROX. 7/8″)
Oxygen Flow Range – 3-10 LPM
Nitrous Oxide Flow Range – 0-7 LPM
Fail-Safe Type – Proportional
Tall Stand Height – 40″- 50″
Legal Disclaimer: The Belmed PC-7 and PC-7C dental flowmeters shall only be used by dentists, physicians, or other practitioners that are qualified and licensed to use nitrous oxide sedation. The Belmed PC-7 and PC-7C dental flowmeters shall be used in conjunction with a scavenging/vacuum system to remove any free floating nitrous oxide (N2O) gas from the ambient environment, as the CDC has reported that exposure to free-floating nitrous oxide gas can have adverse health effects such as dizziness, coughing, shortness of breath, unconsciousness, and infertility.
MGI Inc. (authorized dealer) and Belmed Inc. (manufacturer) shall not be liable for any damage, injury or loss arising out of the use of the product, whether as a result of a defect in the product or otherwise, if prior to such damage, injury or loss, the product was (1) damaged, misused, or misapplied; (2) repaired, altered or modified by persons other than Belmed, Inc. (3) not installed in strict compliance with applicable codes and ordinances; or (4) not installed by Belmed, Inc. or an authorized Belmed, Inc. dealer.
|Nitrous Oxide Flow Range|